This is a step-by-step process to calibrate your NeoFox Oxygen Sensor in both gas and liquid phases. It is important to calibrate your system to meet your conditions.
2-Point Oxygen Calibration for Gas and Liquid Samples
You may also like
Featured on our Thursday's Spotlight Series is our Very own Yvette Mattley's SQUARE ONE repeatability Study. How to Ensure Reliable Absorbance Results. Learn More: https://t.co/k1JmAuasS62
3, 2, 1 - Fire! #OHB aboard! 🔥🚀@oceanoptics and the Human Spaceflight Department of @OHB_SE, under a contract of @esa, are cooperating on the customization of an UV-VIS spectrometer as key non-invasive methods to analyze the samples in-situ. https://t.co/p0r4rKhHy7
SPECTROdemo is launching today at 4:46 P.M on the #CygnusSpacecraft , to test key functions of the spectrometer! #DidYouKnow Performing experiments directly in space offers the great chance to overcome the difficulty to mimic environmental conditions? https://t.co/4sK4GoU5D5